In Ubuntu 14.04 with dispcalGUI 18.104.22.168 with Argyll 1.6.3 and Hugski calibration device, when I start calibration, it displays a window for Interactive setup of display and when I click on Start measure it displays few color blocks and then get stuck on white and beep everything second. Whatever I play with color settings of display the value displayed in calibration window don’t change 🙁
Any ideas what’s wrong ?
This topic was modified on 2015-07-21 15:47:07 by vincen_pujol.
it beeps each time a reading is taken. For the ColorHug, I would recommend to not do any whitepoint adjustment though, it’s not accurate enough on a lot of displays without a custom-made colorimeter correction (this is a known shortcoming of the ColorHug hardware). Cancel the calibration, then set measurement mode to “Auto” and reset the display to factory defaults. Then start the process again.
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.