In Ubuntu 14.04 with dispcalGUI 188.8.131.52 with Argyll 1.6.3 and Hugski calibration device, when I start calibration, it displays a window for Interactive setup of display and when I click on Start measure it displays few color blocks and then get stuck on white and beep everything second. Whatever I play with color settings of display the value displayed in calibration window don’t change 🙁
Any ideas what’s wrong ?
This topic was modified on 2015-07-21 15:47:07 by vincen_pujol.
it beeps each time a reading is taken. For the ColorHug, I would recommend to not do any whitepoint adjustment though, it’s not accurate enough on a lot of displays without a custom-made colorimeter correction (this is a known shortcoming of the ColorHug hardware). Cancel the calibration, then set measurement mode to “Auto” and reset the display to factory defaults. Then start the process again.